|Mounting conditions||Diffuse reflection|
|Reference distance||80 mm|
|Measurement range||Z-axis (height)||±23 mm|
|X-axis (width)||NEAR side||25 mm|
|Reference distance||32 mm|
|Far side||39 mm|
|Light source||Type||Blue semiconductor laser|
|Wavelength||405 nm (visible beam)|
|Laser class||Class 2 Laser Product|
(IEC60825-1, FDA(CDRH) Part 1040.10*1)
|Spot size (reference distance)||Approx. 48 mm ? 48 µm|
|Repeatability||Z-axis (height)||0.5 µm*2*3|
|X-axis (width)||10 µm*2*4|
|Linearity||Z-axis (height)||±0.1% of F.S.*5|
|Profile data interval||X-axis (width)||50 µm|
|Sampling cycle (trigger interval)||Top speed: 125 µs*6|
|Temperature characteristics||0.01% of F.S./°C|
|Environmental resistance||Enclosure rating||IP67 (IEC60529)*7|
|Ambient light||Incandescent lamp: 10,000 lux max.*8|
|Ambient temperature||0 to +45 °C*9|
|Relative humidity||20 to 85 % RH (No condensation)|
|Vibration resistance||10 to 55 Hz, Double amplitude 1.5 mm, 3 hours in each of the X, Y, and Z directions|
|Shock resistance||15 G/6 ms|
|Weight||Approx. 400 g|
|*1 The laser classification for FDA(CDRH) is implemented based on IEC60825-1 in accordance with the requirements of Laser Notice No. 50.|
*2 This value is from a case in which measurement has been performed with a reference distance with 4,096 times of averaging.
*3 The measurement targets are KEYENCE standard targets. This value is from a case in which the average height of the default setting area has been measured in height mode. All other settings are default.
*4 The measurement target is a pin gauge. This value is from a case in which the position of the intersection between the rounded surface of the pin gauge and the edge level has been measured in position mode. All other settings are default.
*5 The measurement targets are KEYENCE standard targets. The profile data is from a case in which measurement has been performed with 64 times of smoothing and 8 times of averaging. All other settings are default.
*6 Sampling frequency depends on the setting of measurement range, binning and parallel imaging.
*7 This value is from a case in which the sensor head cable (CB-B*) or extension cable (CB-B*E) has been connected.
*8 This is the illuminance for the light-receiving surface of the sensor head during white paper measurement when light has been shined onto the white paper.
*9 The sensor head must be mounted on a metal plate for use.
Датчик измерения Keyence LJ-V7080B
Sensor head (Light intensity output type)
Категория: Датчики измерения